The COVID-19 pandemic has continued to cause dramatic shifts in the practice of otolaryngology. Even with standard precautions such as physical distancing and wearing personal protective equipment, aerosol-generating procedures such as nasolaryngoscopy (a commonly performed in-office procedure in which a soft, flexible fiber-scope is passed through the nose and into the throat) and intranasal instrumentation were determined to carry a risk of potential transmission if not adequately protected.
In an effort to mitigate exposure to these airborne particles, researchers from Boston University School of Medicine (BUSM) designed and tested a prototype nasolaryngoscopy hood, worn by the patient that offers safe and effective protection in reducing aerosols exposures.
In order to test the efficacy of the hood, a particle counter was used to calculate the average number of 0.3-mm particles/L detected during various clinical scenarios that included sneezing, nasolaryngoscopy, sneezing during nasolaryngoscopy and topical lidocaine spray administration. Experiments were repeated to compare the effectiveness of the hood versus no protection.
When no patient barrier (hood or mask) was used, a significant increase in aerosols was detected during sneezing, sneezing during nasolaryngoscopy and topical spray administration. With the hood was in place, the level of aerosols returned to baseline levels in each scenario.
“This simple intervention allows patients to undergo routine flexible nasal laryngoscopy, even with topical lidocaine spray administration, with less risk to the provider,” explained corresponding author Christopher Brook, MD, assistant professor of otolaryngology—head and neck surgery at BUSM. “If a patient begins to sneeze during the examination, our data suggest that providers will remain protected through the use of the hood,” he added.
While this study evaluated the efficacy of the hood in the setting of a routine nasolaryngoscopy in reducing aerosol spread, there are other possible applications but also one major barrier to overcome. “For clinical use, either the hood would need to be mass produced to allow for single use, or a safe and effective protocol of cleaning and reusing each hood would need to be established,” said Brook who also is an otolaryngologist at Boston Medical Center.
Way forward in protecting healthcare workers from COVID-19
Michal J. Plocienniczak et al, Evaluating a Prototype Nasolaryngoscopy Hood During Aerosol-Generating Procedures in Otolaryngology, Otolaryngology–Head and Neck Surgery (2020). DOI: 10.1177/0194599820973652
Cost-effective hood reduces aerosol exposures to patients, otolaryngologists (2020, December 23)
retrieved 23 December 2020
This document is subject to copyright. Apart from any fair dealing for the purpose of private study or research, no
part may be reproduced without the written permission. The content is provided for information purposes only.
- COVID case rates hit new high for England, study finds - April 7, 2022
- Govt’s focus on affordable healthcare ensured significant savings for poor, middle class: PM Modi - April 7, 2022
- SRL Diagnostics and Skye Air Mobility collaborate to transport pathology samples using drone logistics - April 6, 2022
- Healthineers sets up new production line of CT scanners in Bengaluru under PLI scheme - April 6, 2022
- Lupin inks licensing pact with Alvion to market drugs in Southeast Asia - April 6, 2022
- Yoga Mahotsav: Ayush Ministry to organise event to demonstrate common yoga on World Health Day - April 6, 2022
- LordsMed forays into the medtech space with launch of health ATMs ‘Lords Sehat’ - April 5, 2022
- ‘Friendly viruses’ can be the next big thing in the history of medical research and more - April 5, 2022
- No setback to Bharat Biotech even as WHO suspends Covaxin UN supply: Sources - April 4, 2022
- Govt panel recommends Serum’s Covovax dose for kids aged 12 and above - April 4, 2022